Author: Lienhard K1, Cabasson A2, Meste O2, Colson SS3.
1University of Nice Sophia Antipolis, CNRS, I3S, UMR7271, Sophia Antipolis, France; University of Nice Sophia Antipolis, LAMHESS, EA 6312, Nice, France; University of Toulon, LAMHESS, EA 6312, La Garde, France. Electronic address: email@example.com. 2University of Nice Sophia Antipolis, CNRS, I3S, UMR7271, Sophia Antipolis, France. 3University of Nice Sophia Antipolis, CNRS, I3S, UMR7271, Sophia Antipolis, France; University of Nice Sophia Antipolis, LAMHESS, EA 6312, Nice, France; University of Toulon, LAMHESS, EA 6312, La Garde, France.
Conference/Journal: J Electromyogr Kinesiol
Date published: 2015 Nov 9
Other: Pages: S1050-6411(15)00195-9 , Special Notes: doi: 10.1016/j.jelekin.2015.10.005 , Word Count: 260
The objective was to investigate the influence of surface electromyography (sEMG) processing methods on the quantification of muscle activity during whole-body vibration (WBV) exercises. sEMG activity was recorded while the participants performed squats on the platform with and without WBV. The spikes observed in the sEMG spectrum at the vibration frequency and its harmonics were deleted using state-of-the-art methods, i.e. (1) a band-stop filter, (2) a band-pass filter, and (3) spectral linear interpolation. The same filtering methods were applied on the sEMG during the no-vibration trial. The linear interpolation method showed the highest intraclass correlation coefficients (no vibration: 0.999, WBV: 0.757-0.979) with the comparison measure (unfiltered sEMG during the no-vibration trial), followed by the band-stop filter (no vibration: 0.929-0.975, WBV: 0.661-0.938). While both methods introduced a systematic bias (P<0.001), the error increased with increasing mean values to a higher degree for the band-stop filter. After adjusting the sEMGRMS during WBV for the bias, the performance of the interpolation method and the band-stop filter was comparable. The band-pass filter was in poor agreement with the other methods (ICC: 0.207-0.697), unless the sEMGRMS was corrected for the bias (ICC⩾0.931, %LOA⩽32.3). In conclusion, spectral linear interpolation or a band-stop filter centered at the vibration frequency and its multiple harmonics should be applied to delete the artifacts in the sEMG signals during WBV. With the use of a band-stop filter it is recommended to correct the sEMGRMS for the bias as this procedure improved its performance.
Copyright © 2015 Elsevier Ltd. All rights reserved.
Band-stop filter; Motion artifacts; Power spectral density; Spectral linear interpolation; Surface electromyography